Calibration Leaks for Sensistor Industrial Hydrogen Leak Detectors
The Easy Way to Test and to Calibrate - For Maximum Accuracy
To be able to correctly accept/reject test objects you need to measure to a set standard. You also need to calibrate your leak detector against a reliable reference leak. INFICON reference leaks for Sensistor industrial hydrogen leak detectors cover a wide range of leak rates to suit your specific application and include bigger leaks (Types A-C) and smaller leaks (Types E and G). Leak Type A is intended for accumulation testing only.
All leaks are traceable to NIST, NMIJ, NPL, PTB, etc., through the Mutual Recognition Arrangement of the BIPM.
TYPES AND CONNECTIONS
A, B and C
Sintered stainless steel leaks
Target flow between 5x10-2 and 5x10-4 (Air)
E and G
Crimped metal capillary leaks
Target flow between 7x10-5 and 2x10-5 (5%H2/95%N2)
- Suitable for industrial applications
- Easy to use
- Available in different leak rates
- Traceable to NIST, NMIJ, NPL, PTB, etc.
- Test and Calibration of Sensistor Industrial Hydrogen Leak Detectors
|Calibration Leaks for Sensistor Industrial Leak Detectors|
|590-420||Calibration Leak Type A, 5x10E-2 mbarl/s @1bar|
|590-421||Calibration Leak Type B, 5x10E-3 mbarl/s @1bar|
|590-422||Calibration Leak Type C, 5x10E-4 mbarl/s @1bar|
|590-427||Calibration Leak Type E, 10g/a|
|590-429||Calibration Leak Type G, 3g/a|